FACTOID # 25: If you're tired of sitting in traffic on your way to work, move to North Dakota.

 Home Encyclopedia Statistics States A-Z Flags Maps FAQ About

 WHAT'S NEW RELATED ARTICLES People who viewed "Anemometer" also viewed:

SEARCH ALL

Search encyclopedia, statistics and forums:

(* = Graphable)

Encyclopedia > Anemometer
A hemispherical cup anemometer of the type invented in 2000 by John Thomas Romney Robinson

Cup-type anemometer with vertical axis and turnabout counter located at the Dübendorf museum of military aviation

Anemometers can be divided into two classes: those that measure the velocity of the wind, and those that measure the pressure of the wind, but as there is a close connection between the pressure and the velocity and a suitable anemometer of either class will give information about both these quantities. Image File history File linksMetadata Anemometer_dubendorf. ... Image File history File linksMetadata Anemometer_dubendorf. ... DÃ¼bendorf is a suburb of ZÃ¼rich in Switzerland with a population of about 21,106 (in 1990). ... In physics, velocity is defined as the rate of change of displacement or the rate of displacement. ...

### Cup anemometers

The simplest type of anemometer is the cup-anemometer, invented (1846) by Dr. John Thomas Romney Robinson, of Armagh Observatory. It consisted of four hemispherical cups each mounted on one end of four horizontal arms, which in turn were mounted at equal angles to each other on a vertical shaft. The air flow past the cups in any horizontal direction turned the cups in a manner that was proportional to the wind speed. Therefore counting the turns of the cups over a set time period produced the average wind speed for a wide range of speeds. On an anemometer with four cups it is easy to see that since the cups are arranged symmetrically on the end of the arms, the wind always has the hollow of one cup presented to it and is blowing on the back of the cup on the opposite end of the cross. Studies of the forces involved show that the force is greater on the cup side of the device and the net force causes the cups to spin, but in this case the balance of forces is not obvious. John Thomas Romney Robinson (April 23, 1792 - February 28, 1882) was an Irish astronomer and physicist. ... Armagh Observatory is a modern astronomical research institute with a rich heritage, based in Armagh, Northern Ireland. ... The term hemisphere is used in three different meanings: one-half of the Earth (or other planetary or stellar body; see also New World and Old World) Eastern Hemisphere and Western Hemisphere. ...

Unfortunately, when Robinson first designed his anemometer, he stated that no matter what the size of the cups or the length of the arms, the linear speed of the cups always moved with one-third of the speed of the wind. This result was apparently confirmed by some early independent experiments, but it is very far from the truth. It was later discovered that the actual relationship between the speed of the wind and that of the cups, called the anemometer factor, depended on the dimensions of the cups and arms, and may have a value between two and a little over three. This had the result that wind speeds published in many official 19th century publications were often in error.

The three cup anemometer developed by the Canadian John Patterson in 1926 and subsequent cup improvements by Brevoort & Joiner of the USA in 1935 led to a cupwheel design which was linear and had an error of less than 3% up to 60 mph. Patterson found that each cup produced maximum torque when it was at 45 degrees to the wind flow. The three cup anemometer also had a more constant torque and responded more quickly to gusts than the four cup anemometer.

The three cup anemometer was further modified by the Australian Derek Weston in 1991 to measure both wind direction and wind speed. Weston added a tag to one cup, which causes the cupwheel speed to increase and decrease as the tag moves alternately with and against the wind. Wind direction is calculated from these cyclical changes in cupwheel speed, while wind speed is as usual determined from the average cupwheel speed.

A windmill style of anemometer

Image File history File links Download high resolution version (1824x1208, 989 KB) Description: Project Vortex mesonet vehicle aerovane for wind speed and direction. ... Image File history File links Download high resolution version (1824x1208, 989 KB) Description: Project Vortex mesonet vehicle aerovane for wind speed and direction. ...

### Windmill anemometers

The other forms of mechanical velocity anemometer may be described as belonging to the windmill type. In the Robinson anemometer the axis of rotation is vertical, but with this subdivision the axis of rotation must be parallel to the direction of the wind and therefore horizontal. Furthermore, since the wind varies in direction and the axis has to follow its changes, a wind vane or some other contrivance to fulfill the same purpose must be employed. An aerovane combines a propeller and a tail on the same axis to obtain accurate and precise wind speed and direction measurements from the same instrument. In cases where the direction of the air motion is always the same, as in the ventilating shafts of mines and buildings for instance, wind vanes, known as air meters are employed, and give most satisfactory results. A Dutch tower windmill, sporting sails, surrounded by tulips A windmill is an engine powered by the wind to produce energy, often contained in a large building as in traditional post mills, smock mills and tower mills. ... Weather vane Weather cock Aerovane A weather vane, also called a wind vane, is a movable device attached to an elevated object such as a roof for showing the direction of the wind. ... An anemometer is a device for measuring the velocity or the pressure of the wind, and is one instrument used in a weather station. ...

### Hot-wire anemometers

Hot wire anemometers use a very fine wire (on the order of several micrometers) heated up to some temperature above the ambient. Air flowing past the wire has a cooling effect on the wire. As the electrical resistance of most metals is dependent upon the temperature of the metal (tungsten is a popular choice for hot-wires), a relationship can be obtained between the resistance of the wire and the flow velocity.[1] For other uses, see Tungsten (disambiguation). ...

Several ways of implementing this exist, and hot-wire devices can be further classified as CCA (Constant-Current Anemometer), CVA (Constant-Voltage Anemometer) and CTA (Constant-Temperature Anemometer). The voltage output from these anemometers is thus the result of some sort of circuit within the device trying to maintain the specific variable (current, voltage or temperature) constant.

Additionally, PWM (Pulse-width modulation) anemometers are also used, wherein the velocity is inferred by the time length of a repeating pulse of current that brings the wire up to a specified resistance and then stops until a threshold "floor" is reached, at which time the pulse is sent again. Pulse-width modulation of a signal or power source involves the modulation of its duty cycle, to either convey information over a communications channel or control the amount of power sent to a load. ...

Hot-wire anemometers, while extremely delicate, have extremely high frequency-response and fine spatial resolution compared to other measurement methods, and as such are almost universally employed for the detailed study of turbulent flows, or any flow in which rapid velocity fluctuations are of interest.

### Laser Doppler anemometers

Drawing of a laser anemometer. The laser is emitted (1) through the front lens (6) of the anemomerter and is backscattered off the air molecules (7). The backscattered radiation (dots) re-enter the device and are reflected and directed into a detector (12).

Laser Doppler anemometers use a beam of light from a laser that is split into two beams, with one propagated out of the anemometer. Air molecules near where the beam exits reflect, or backscatter, the light back into a detector, where it is measured relative to the original laser beam. When the air molecules are in motion, they produce a doppler shift in the laser light, which is used to calculate the speed of the molecules, and therefore the air around the anemometer.[2]
Image File history File links Download high-resolution version (1186x588, 207 KB) File links The following pages on the English Wikipedia link to this file (pages on other projects are not listed): Anemometer ... Image File history File links Download high-resolution version (1186x588, 207 KB) File links The following pages on the English Wikipedia link to this file (pages on other projects are not listed): Anemometer ... For other uses, see Laser (disambiguation). ... The Doppler effect is the apparent change in frequency or wavelength of a wave that is perceived by an observer moving relative to the source of the waves. ...

3D ultrasonic anemometer

### Sonic anemometers

Two dimensional (wind speed and wind direction) sonic anemometers are used in applications such as small weather stations, ship navigation, wind turbines and aviation.

## Pressure anemometers

The first designs of anemometers which measure the pressure were divided into plate and tube classes.

### Plate anemometers

These are the earliest anemometers and are simply a flat plate suspended from the top so that the wind deflects the plate. They date from the Italian Leon Battista Alberti in about 1450 and Robert Hooke in 1664. Later versions of this form consisted of a flat plate, either square or circular, which is kept normal to the wind by a wind vane. The pressure of the wind on its face is balanced by a spring. The compression of the spring determines the actual force which the wind is exerting on the plate, and this is either read off on a suitable gauge, or on a recorder. Instruments of this kind do not respond to light winds, are inaccurate for high wind readings, and are slow at responding to variable winds. Plate anemometers have been used to trigger high wind alarms on bridges.

### Tube anemometers

Helicoid propeller anemometer incorporating a wind vane for orientation.

James Lind's anemometer of 1775 consisted simply of a glass U tube containing liquid, a manometer, with one end bent in a horizontal direction to face the wind and the other vertical end remains parallel to the wind flow. Though the Lind was not the first it was the most practical and best known anemometer of this type. If the wind blows into the mouth of a tube it causes an increase of pressure on one side of the manometer. The wind over the open end of a vertical tube causes little change in pressure on the other side of the manometer. The resulting liquid change in the U tube is an indication of the wind speed. Small departures from the true direction of the wind causes large variations in the magnitude. Image File history File links No higher resolution available. ... Image File history File links No higher resolution available. ... The helicoid is one of the first minimal surfaces discovered. ... A weather vane, also called a wind vane, is a movable device attached to an elevated object such as a roof for showing the direction of the wind. ...

The highly successful metal pressure tube anemometer of William Henry Dines in 1892 utilized the same pressure difference between the open mouth of a straight tube facing the wind and a ring of small holes in a vertical tube which is closed at the upper end. Both are mounted at the same height. The pressure differences on which the action depends are very small, and special means are required to register them. The recorder consists of a float in a sealed chamber partially filled with water. The pipe from the straight tube is connected to the top of the sealed chamber and the pipe from the small tubes is directed into the bottom inside the float. Since the pressure difference determines the vertical position of the float this is a measure of the wind speed.

The great advantage of the tube anemometer lies in the fact that the exposed part can be mounted on a high pole, and requires no oiling or attention for years; and the registering part can be placed in any convenient position. Two connecting tubes are required. It might appear at first sight as though one connection would serve, but the differences in pressure on which these instruments depend are so minute, that the pressure of the air in the room where the recording part is placed has to be considered. Thus if the instrument depends on the pressure or suction effect alone, and this pressure or suction is measured against the air pressure in an ordinary room, in which the doors and windows are carefully closed and a newspaper is then burnt up the chimney, an effect may be produced equal to a wind of 10 mi/h (16 km/h); and the opening of a window in rough weather, or the opening of a door, may entirely alter the registration.

While the Dines anemometer had an error of only 1% at 10mph it did not respond very well to low winds due to the poor response of the flat plate vane required to turn the head into the wind. In 1918 an aerodynamic vane with eight times the torque of the flat plate overcame this problem.

### Effect of density on measurements

In the tube anemometer the pressure is measured, although the scale is usually graduated as a velocity scale. In cases where the density of the air is significantly different from the calibration value (as on a high mountain, or with an exceptionally low barometer) an allowance must be made. Approximately 1½% should be added to the velocity recorded by a tube anemometer for each 1000 ft (5% for each kilometer) above sea-level.

 Nautical Portal
• Anemoscope, ancient device for measuring or predicting wind direction or weather
• Weather vane, device for indicating wind direction
• Windsock, device for indicating wind speed and direction

Image File history File links No higher resolution available. ... An anemoscope is an obsolete machine invented to show the direction of the wind, or to foretell a change of wind direction or weather. ... Weather vane Weather cock Aerovane A weather vane, also called a wind vane, is a movable device attached to an elevated object such as a roof for showing the direction of the wind. ... A windsock is a large, conical, open-ended tube designed to indicate wind direction and relative wind speed. ...

## References

1. ^ Hot-wire Anemometer explanation. eFunda. Retrieved on September 18, 2006.
2. ^ Iten, Paul D. (June 29, 1976). Laser doppler anemometer. United States Patent and Trademark Office. Retrieved on September 18, 2006.

Meteorological Instruments, W.E. Knowles Middleton and Athelstan F. Spilhaus, Third Edition revised, University of Toronto Press, Toronto, 1953 Supporters contend that the Eleventh Edition of the EncyclopÃ¦dia Britannica (1910-1911) represents the sum of human knowledge at the beginning of the 20th century; indeed, it was advertised as such. ... is the 180th day of the year (181st in leap years) in the Gregorian calendar. ... Year 1976 Pick up sticks(MCMLXXVI) was a leap year starting on Thursday (link will display full calendar) of the Gregorian calendar. ...

Invention of the Meteorological Instruments, W.E. Knowles Middleton, The John Hopkins Press, Baltimore, 1969

Results from FactBites:

 Britain.tv Wikipedia - Anemometer (1504 words) An anemometer is a device for measuring the velocity or the pressure of the wind, and is one instrument used in a weather station. Anemometers may be divided into two classes: those that measure the velocity of the wind, and those that measure the pressure of the wind, but as there is a close connection between the pressure and the velocity, a suitable anemometer of either class will give information about both these quantities. Hot-wire anemometers, while extremely delicate, have extremely high frequency-response and fine spatial resolution compared to other measurement methods, and as such are almost universally employed for the detailed study of turbulent flows, or any flow in which rapid velocity fluctuations are of interest.
 Anemometer (glossary definition) (157 words) Cup anemometer is used to measure the wind speed from the speed of rotation of a windmill which consist of 3 or 4 hemispherical or conical cups, each fixed to the ends of horizontal arms attached to a vertical axis. Counting anemometer has cups or a fan whose rotation is transmitted to a technical counter which integrates directly the air movement speed. Pressure tube anemometer (Dines anemometer) is an instrument which derives wind speed from measurements of the dynamic wind pressures.
More results at FactBites »

Share your thoughts, questions and commentary here